Keysight N7810A PSA Series
Calibration Application Software
Fixed a localization issue with Frequency
Response Below 300kHz.
Fixed an ETE option bug in the Noise
Sidebands test where PSG option UNY (low noise source) was
omitted in error.
Fixed an IF Amplitude Ripple report bug
where the legacy sub-report template was being used in
Updated the equipment database to include
the 87300C Coupler to fix a runtime error in the Gain
Updated the Frequency Response 300 kHz to
3 GHz test so thepower level for UUTs with preamp option 1DS
or option 110 matches the TSD @ -38dBm.
© Keysight Technologies 2000-2017