Added support for the 33611A/12A/21A/22A
Waveform Generators in
three PV tests: Audio Amplitude
Accuracy, Frequency Response
Below 300 kHz and Preselector
Tune Out Accuracy.
Added support for the 34470A Digital
Multimeter in two PV tests:
Frequency Response Below 300 kHz
and Preselector Tune Out Accuracy.
Fixed the Calibration Constant Reset
Utility on models E4446/7/8A.
The utility will now correctly
reset the calibration constants
to the factory default
The recommended variant name is now
Added a new Noise Sidebands test that
tests both near and far
offsets. Two equipment solutions are
available: Wenzel (preferred)
and PSG/Notch (requires a low
Note: This test is included in the Factory
Recommended variant only.
It replaces the Noise Sidebands >
50 kHz Offsets and Noise Sidebands
< 50 kHz Offsets tests in
the Factory Recommended variant; both
are still included in
the Normal variant.
Noise Sidebands (Wenzel) no longer
requires a 6dB Fixed Attenuator
and a 50 ohm terminator to be
mapped, and neither will appear on the
report. However the
6dB attenuator is still required as part of
the test setup
and appears in the connection setup prompt.
Sidebands > 50 kHz Offsets (Wenzel) still requires both
mapped and will show up on the report.
Fixed an uncertainty bug for the 3458A
Digital Multimeter related to
TCAL: if the internal
temperature drift is greater than +/-5 degrees C,
spec is applied to the DMM, which will change the uncertainty.
The installation package now supports
Added support for the PSG E8257N.
Other miscellaneous bug fixes.